Our most advanced endocutter technology – ECHELON FLEX™ Powered ENDOPATH® Stapler
Design refined by experience
By collaborating with leaders in the surgical community, the development of endocutters is grounded in Halsted's principles of surgery – the importance of hemostasis, the gentle handling of tissue, and the application of appropriate tissue compression for safe and effective surgery.
Now experience less tissue slippage during firing with the ECHELON FLEX™ GST System*ECHELON FLEX™ GST System
Enables the most precise placement on fragile vesselsECHELON FLEX™ Powered Vascular Stapler
Stability for potentially less trauma on vital thoracic structures† and on thick tissue‡ECHELON FLEX™ Powered ENDOPATH® Stapler
One-handed articulation plus hemostasis in a wide range of tissue thicknessECHELON FLEX™ ENDOPATH® Staplers
- * Benchtop testing in porcine stomach tissue. Mean tissue movement from after clamping on tissue to after firing ECHELON FLEX™ Powered Plus Stapler (PSEE60A) and ECHELON Reload with GST vs ENDO GIA™ ULTRA Handle (EGIAUSTND) and Endo GIA™ Reload with Tri-Staple™ Technology at 1.5, 2.5, 3.3 and 4.0mm tissue thicknesses (1.5mm; GST60B 1.067mm vs EGIA60AMT 2.452mm p<0.001; 2.5mm: GST60G 1.148mm vs EGIA60AMT 3.261mm p<0.001; 3.3mm: GST60T 0.642mm vs EGIA60AMT 4.806mm p<0.001; 4.0mm: GST60T 0.654mm vs EGIA60AXT 5.116mm p<0.001).
- † Benchtop testing on porcine carotid arteries. Surgeons (n=24) fired each instrument / reload once: PSE45A / ECR45WG and EGIAUSTND / EGIA45AVM. Distal tip motion measurement during the firing cycle showed a median reduction of 83% and a range of reduction of 53% to 96% in tip movement of PSE45A/ECR45W vs. EGIAUSTND / EGIA45AVM .
- ‡ Benchtop testing on porcine stomach tissue. Surgeons (n=19) fired each instrument/ reload once: PSE60A/ ECR60G, 030449/ 030459, and EGIAUSTND/ EGIA60AMT. Distal tip motion measurement during the firing cycle showed a median of 88% and a range of reduction of 71% to 95% in tip movement of PSE60A/ ECR60G vs. the other two devices.