Our most advanced endocutter technology – ECHELON FLEX™ Powered ENDOPATH® Stapler
Design refined by experience
By collaborating with leaders in the surgical community, the development of endocutters is grounded in Halsted's principles of surgery – the importance of hemostasis, the gentle handling of tissue, and the application of appropriate tissue compression for safe and effective surgery.Stapling Product Reference Chart picture_as_pdf
Now experience less tissue slippage during firing with the ECHELON FLEX™ GST System*ECHELON FLEX™ GST System
Enables the most precise placement on fragile vesselsECHELON FLEX™ Powered Vascular Stapler
Stability for potentially less trauma on vital thoracic structures† and on thick tissue‡ECHELON FLEX™ Powered ENDOPATH® Stapler
One-handed articulation plus hemostasis in a wide range of tissue thicknessECHELON FLEX™ ENDOPATH® Staplers
- * Benchtop testing in porcine stomach tissue. Mean tissue movement from after clamping on tissue to after firing ECHELON FLEX™ Powered Plus Stapler (PSEE60A) and ECHELON Reload with GST vs ENDO GIA™ ULTRA Handle (EGIAUSTND) and Endo GIA™ Reload with Tri-Staple™ Technology at 1.5, 2.5, 3.3 and 4.0mm tissue thicknesses (1.5mm; GST60B 1.067mm vs EGIA60AMT 2.452mm p<0.001; 2.5mm: GST60G 1.148mm vs EGIA60AMT 3.261mm p<0.001; 3.3mm: GST60T 0.642mm vs EGIA60AMT 4.806mm p<0.001; 4.0mm: GST60T 0.654mm vs EGIA60AXT 5.116mm p<0.001).
- † Benchtop testing on porcine carotid arteries. Surgeons (n=24) fired each instrument / reload once: PSE45A / ECR45WG and EGIAUSTND / EGIA45AVM. Distal tip motion measurement during the firing cycle showed a median reduction of 83% and a range of reduction of 53% to 96% in tip movement of PSE45A/ECR45W vs. EGIAUSTND / EGIA45AVM .
- ‡ Benchtop testing on porcine stomach tissue. Surgeons (n=19) fired each instrument/ reload once: PSE60A/ ECR60G, 030449/ 030459, and EGIAUSTND/ EGIA60AMT. Distal tip motion measurement during the firing cycle showed a median of 88% and a range of reduction of 71% to 95% in tip movement of PSE60A/ ECR60G vs. the other two devices.